Lab supply | First online laboratory shop, and selling where new features were introduced. practice. Instrumental factor must be checked. Chema Diagnostica will not be responsible for instrumental bad programming. Mindray BAA. Mindray) owns the intellectual property rights to this Mindray product and this manual. This manual may refer to information protected by copyrights or patents.
|Published (Last):||11 November 2005|
|PDF File Size:||9.70 Mb|
|ePub File Size:||11.89 Mb|
|Price:||Free* [*Free Regsitration Required]|
Remove the red ring on the used tubing and install them on the new tubing.
Troubleshooting This chapter presents all warning messages and recommended corrective actions, which should be taken in time once any kindray occurs. Preventing Electric Shock Please observe the following instructions to prevent electric shock.
Replacing The Tubing Ba-88w the system. To print the manual completely, please, download it. Introduction of the Modules The hardware system of the BAA is divided into the following modules or boards: NOTE Wear clean cotton gloves, when replacing the filter or lens.
NOTE When replacing the lamp assembly, do not touch the surface of the bulb to avoid contamination. When replacing the optical components, take care to protect the components. Page 66 After unplugging the adapters, pinch the buckles on the pump head and detach the pump shell. Find more suppliers in Bengaluru Clinical Analyzer in Bengaluru.
Figure Structure of the CPU extension board 4. Connect the thicker end of the new tube to the inlet metal tube and then guide the tube through the deflection tube. Semi Auto Analyser in Bengaluru. Get Best Price Request a quote. Page 26 The replacement of lamp assembly is shown in the following b-88a. Our company is a sole proprietorship based firm.
Pull out the tubing that goes through the backboard of the analyzer for mm until the adapter is exposed. Service and Maintenance To ensure reliability, good performance and service life of the system, regular maintenance is required. Date of Manufacture Manufacturer CE marking. Page mindra This manual is geared for service personnel authorized by Mindray. Page 36 Cover Recorder Aspirate button movement lamp switch replacement window Hole for holding flow cell temporarily Enclosure fixing screw Aspirate button hole Enclosure Touchscreen Touch screen supporting board 3.
Detection Of The Input Signal 4. Replacement of the peristaltic pump should be followed with flow volume calibration this is not for system configured minddray cuvette. Replacing Lamp Connect the thicker end of the new tube to the inlet metal tube and then guide the tube through the deflection tube.
NOTE If either of the temperature sensor or analogue board is damaged, both of the two should be replaced by one assembly at the same time. BA A is a Semi – Auto chemistry analyzer and a computer in one unit. Of the 11 screws, 3 are located at the back of the system and the remaining which baa-88a be dismounted directly are located under the enclosure.
Microplate reader MRA Mindray. Unscrew screws remove replacement window cover Unscrew the fastening screw of the lamp base if it is difficult to dismount, some tools like the clipper might be used.
Click start and a dialog box notifying the user to shut down lower computer communication will be displayed.
Figure System structure of the software 5. Page 11 Treating Waste Liquids Please observe the following instructions minxray prevent environmental pollution and personal mindrag caused by waste. Connection Of The Hardware System 4. Factory price Mindray touch screen baa semi-auto chemistry analyzer with User-friendly interface. Product Specification Automation Grade Manual.
Minfray Burkart leaves Entscheiderfabrik The last day of the fair at this year’s Medica was also the last Modification on the screen. Do not stare into the lamp when the system is in operation. Bi-chromatic experiments for kinetics, fixed-time and end mkndray methodologies are supported efficiently. System Specifications System specifications Dimension: Enclosure Assembly refrigeration plate, lamp cable and the connection cable of analogue board. User-friendly interface and multi-language operation software.
If obvious blow-up is found, skewed installation of the lens might be the cause and the readjustment might be needed. Page 10 Preventing Personal Injury Caused by Photometer Lamp Please observe the following instructions to prevent personal injury caused by photometer lamp. NOTE The service life of the pump tubing is 18 month.
If not, execute the CPU board software upgrading again. The configuration procedures are shown as follows: This Chemistry Analyzer is semi ba-88z clinical and semi-automatic. Page 58 It will take about 15 minutes for the system to initialize, then the interface to detect the upgrading program on the U minday will be displayed.
The use of this symbol indicates that this product should not be treated as household waste. Replacing Cpu Board Pull the wire out and you can see the wire joint connecting the lamp assembly, with one end connecting to the lamp and the other end connecting to the inside of the instrument.
Chemistry Analyzers in Bengaluru. Semi auto chemistry analyzer. Page 29 surface of the lamp base. Pinch the two buckles on the left and right side of the pump shell and detach the pump mindtay carefully.